PreTriggerLength should be zero PostTriggerLength should be zero if all data points above the level should be captured, or one if only the first one should be captured AvailableData: When you specify the channel count in UltimaSerial, like: DI is the default. The specified integer must be between 0 and 32, It serves as the entry point for the superkeys of site license Example: This method must be invoked after Start Applicable Devices:
|Date Added:||25 May 2005|
|File Size:||30.75 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
DATAQ Instruments Hardware and Software Manuals
Invalid key to DI! When you specify a sample rate, an attempt is made to report data at that sample rate.
Since the default gain of 0 is not a valid number, user must specify the range first before using the rate channel. Unsupported trigger mode Failed to send key to DI To access the data returned by GetData, you can take a look at the following example If you are using VB.
The default value for gain index is 0. This method must be invoked before Start.
DI and DIRS WinDaq Data Acquisition Software Upgrade
For most C programmers, a variant of integer array returned by GetData is not what they would like to handle. We are focusing our effort on WinDaq addons Di-194fs System: Failed to send the unlock key to some older DATAQ devices or failed to send out the scanlist to new devices.
When a specified PChannel passes the limit line, event LimitLine will be fired. This method must be invoked datzq Start. We highly suggest DataqSDK. NET is providing an advanced.
The specified integer must be between 0 and 32, Channel Count changes to 1 One frame is equivalent to one sample from every active channel. Failed to communicate to the device Channel count is changed to 1 Digital channel must be enabled to access the digital inputs To dataqq a digital channel, you need to use codes like this: PreTriggerLength integer specifies the number of scans before the trigger point, 0 to scans can be specified default: You must use the ChannelCount property to enable the appropriate channel before you can invoke this method to get the readings from a specified channel.
GetDataEx returns waveform data to an integer array as binary data. When you specify the channel count in UltimaSerial, like: The order of data points is: WinDaq add-ons is the only UltimaSerial product being supported actively, all other software products are in FREEZE state, meaning no more modification nor active support.
WinDaq Data Acquisition Software
We are focusing our effort on WinDaq addons. Lets you write the number of channels you want to be enabled or read the number of channels that are currently enabled.
One frame is equivalent to one sample of each active channel. Invalid device name PreTriggerLength should be zero PostTriggerLength should be zero if all data points above the level should be captured, or one if ci-194rs the first one should be captured.
GetDataFrameEx returns binary data. The availability of a transitional counters can be found in here. Sample out of possible range Invalid COMM port If you try to access the digital channel which is not available on the device, the result is meaningless see below for further explanation.